2016-03-17

SeisTW 2.4.2.2609 release with the following change:

- The two reference channels are no longer used in Roll-along mode.

- In the trig setup menu, the trig input level is no longer reset to 1 when chosing trig input mode "channel".

- Fixed bug with geophone coordinates in SEG2 file.

Installation:

Before starting the setup program, make sure the following processes are stopped using the Windows Task Manager in the following order: SeisTW, OpenTFTPServerMT, XYNTService, primary and waitprimary. The setup program will ask for the number of available measurement channels. Unfortunately this information is not carried over from the previous installation and needs to be entered manually. Remember that there are always two reference channels included in the sum. If installing to a twelve channel instrument, choose fourteen channels.

After installation has completed, make sure the instrument is completely powered off and then on again. This might include removing the connector to the internal battery if present.